|Subject:||Re: LF: PA0SE Field Strength Meter for 137 kHz|
|From:||"Dick Rollema" <[email protected]>|
|Date:||Tue, 25 Nov 2003 11:29:32 +0100|
To All from PA0SE,|
Many thanks to Stewart and several others who explained what went wrong with my e-mail for some subscribers to the reflector.
I emphasize that the formula can only be used within distances from the transmitter where the field strength decreases in a linear way with distance and attenuation due to absorption by the ground can be neglected.
73, Dick, PA0SE
At 07:17 24-11-03 -0800, you wrote:
Hi Eric and all,
|<Prev in Thread]||Current Thread||[Next in Thread>|
|Previous by Date:||RE: LF: Start rsgb_lf_group? 160-190kHz QRP transmitters in America (Intern), jan-martin . noding|
|Next by Date:||LF: Re: Please forward this mail to Andrew Talbot... [Lf] Frequency standards for LF. The next generation], Alberto di Bene|
|Previous by Thread:||Re: LF: PA0SE Field Srength Meter for 137 kHz, Dave Sergeant|
|Next by Thread:||LF: Start rsgb_lf_group?, Ed Lesnichy|
|Indexes:||[Date] [Thread] [Top] [All Lists]|