Alberto di Bene schrieb:
About Uwe's question, frankly I have no answer... Uwe, how does that level
behave when no
program (neither Argo nor Speclab) is running ? That could give some hints.
73 Alberto I2PHD
Hi Alberto es Wolf,
tks for the answers.
in case neither Argo nor SpecLab is running the level meter
of the PCI 64V-mixer is dark like a night in winter with rain clouded
sky in the northern part of the northern hemisphere far off civilization,
ie no reaction.
spectran does "cooperate" with the PCI 64V, but not fully: the level
meter runs but the input signal takes mysterious tracks and does not
come to the surface agn. the signal hits the WAVE bus instead of the
LINE (or the MIKE).if moving the level "knob" of Spectran by mouse click
the WAVE knob corresponds.
the level meter of the SB16 mixer is switched off and is off screen
when running Argo or Spectran. when running SpecLab the level meter is
switched off but still on screen. in case all programs are off
the level meter is running.