|Subject:||Re: LF: KPTF|
|From:||Markus Vester <[email protected]>|
|Date:||Fri, 23 Nov 2018 23:00:39 +0000 (UTC)|
|Dkim-signature:||v=1; a=rsa-sha256; c=relaxed/relaxed; d=aol.com; s=a2048; t=1543014043; bh=3x1f6nsINcZeaSMyj0LThBtRKDZ3TaoLqDVoI6Um/b0=; h=Date:From:To:Subject:References:From:Subject; b=P83Z4u7qxpT5UR1dQ0UZXeuMm+rI28iIaJnlnOibJlKcjqb4vvWe78obIT/lstu4gthXty+2tOKf1OEFAUEermyeQ4ESdnE+LLRGnKmdHTXfJ7IcjHiJruIM7MMy3UYLcg9vMokDw9IP4JZSgMokaBxWppLxjFtkDvjFTeayurtApRH38ZQuWEG35Uu0BdMe+olKCwGgQminh2y1Ex3/t77tHHcSFO0DFNqW4siLpmNTToFkx44yyv9K8UbaUcxotKcMe+quE4R1GF07JWEEsaFpz8WkKZ5w82ZXFahVlygSqq6/dp4sOqGokKummkwiP/g3ou41VKciOVGoz46e6A==|
that's 1pps leakage from the other stereo channel. Since a couple of weeks, the E-field probe has intermittently lost ELF sensitivity, presumably due to moisture or perhaps being inhabited by bugs and spiders. Now the real ZEVS is absent and the noise is so much attenuated that the leakage becomes visible. In the faster image you can actually see three lines at 1 Hz intervals.
All the best,
Von: Edgar <[email protected]>
An: [email protected] <[email protected]>
Verschickt: Fr, 23. Nov. 2018 23:45
Betreff: LF: KPTF
The phase trace on this image is showing a constant positive slope, what is it measuring?
Is it the difference between the 81.99999 Hz Spectrum Lab signal generator signal and the 82 Hz signal from the PPS?